Design of Radiography Film Dryer Machine Using Timer
- 10.2991/apr.k.220503.016How to use a DOI?
- Dryer Machine; Radiography; Design
Drying is the final step of the film processing cycle which aims to remove the water present in the emulsion layer. The result of the film processing is an undamaged emulsion, free from dust particles, crystal deposits, stains, and artifacts. This research aims to make radiography film dryer with a timer. The stages in this research consist of making image designs and selecting tools and materials, the step of assessing the design results of the component system, the final step of qualitative assessment in the form of function tests and verification of the results using machine. Radiography film dyer machine that have been made with length 90 cm, width 20 cm, height 60 cm, with a maximum capacity of 6 films of 35 cm x 40 cm size. Based on the test results, the temperature of the drying film machine is 30°C - 40°C with 1 radiographic film drying faster, the time is approximately 20 minutes. The model of automatic film dryer that is made is still simple, but it can be used to dry films faster than before.
- © 2022 The Authors. Published by Atlantis Press International B.V.
- Open Access
- This is an open access article distributed under the CC BY-NC 4.0 license.
Cite this article
TY - CONF AU - Efita Pratiwi Adi AU - Anita Nur Mayani AU - Hendri Ermawan PY - 2022 DA - 2022/05/25 TI - Design of Radiography Film Dryer Machine Using Timer BT - Proceedings of the Soedirman International Conference on Mathematics and Applied Sciences (SICOMAS 2021) PB - Atlantis Press SP - 74 EP - 77 SN - 2352-541X UR - https://doi.org/10.2991/apr.k.220503.016 DO - 10.2991/apr.k.220503.016 ID - Adi2022 ER -