Measurement on Surface Temperature Error of Fiber Bragg Grating Temperature Sensor
- 10.2991/asei-15.2015.264How to use a DOI?
- Fiber Bragg grating, temperature sensor, measurement error, compensation.
Fiber Bragg grating temperature sensor has the characteristics of strong anti-jamming capability, distributed measurement which has a good application prospect in all kinds of industrial fields. But in the actual measurement of surface temperature, error caused by temperature gradient caused of the FBG is often ignored. In this paper, researches on finite element and experiments were proposed. The influence factor, and the linear compensation of FBG temperature sensor were found to compensate the property of FBG temperature sensor. The results showed the measurement error ranged from 4 degree centigrade to 14 degree centigrade, which can improve the measurement accuracy of the temperature sensor.
- © 2015, the Authors. Published by Atlantis Press.
- Open Access
- This is an open access article distributed under the CC BY-NC license (http://creativecommons.org/licenses/by-nc/4.0/).
Cite this article
TY - CONF AU - Yi Xie AU - Guofeng Ma AU - Yuan Cheng PY - 2015/05 DA - 2015/05 TI - Measurement on Surface Temperature Error of Fiber Bragg Grating Temperature Sensor BT - Proceedings of the 2015 International conference on Applied Science and Engineering Innovation PB - Atlantis Press SP - 1340 EP - 1343 SN - 2352-5401 UR - https://doi.org/10.2991/asei-15.2015.264 DO - 10.2991/asei-15.2015.264 ID - Xie2015/05 ER -