Wireless Graphene Nanowires QCM Ethanol Vapor Sensor Network Based on ZigBee Protocol
- https://doi.org/10.2991/tmcm-15.2015.30How to use a DOI?
- QCM(quartz crystal microbalance) sensor; ZigBee; graphene nanowires; sensor network
In this paper, a ZigBee wireless ethanol vapor sensor system has been presented. This system is based on quartz crystal microbalance (QCM) coated with graphene nanowires as sensing element. This system is a QCM ethanol vapor sensor network, and we use star network to construct it. It consists of two parts: one is the sensor transmitter module witch consist of lots sensor transmitter; the other is a network center. Every sensor transmitter is a ZigBee coordinator linked with grapheme-nanowires-modified QCM ethanol vapor sensor, and it is responsible to acquire information from QCM sensor and transmit information to network center. The network center constructed with ZigBee coordinator also but linked with computer, and it is responsible to processing and controlling. It is indicated form experimental results that the stability, sensitivity and reproducibility of this wireless sensing system are quite good. The proposed system is low-cost and real-time, and it can not only be used to monitor ethanol vapor but also detect the polluted gas in family and harsh environment.
- © 2015, the Authors. Published by Atlantis Press.
- Open Access
- This is an open access article distributed under the CC BY-NC license (http://creativecommons.org/licenses/by-nc/4.0/).
Cite this article
TY - CONF AU - Jiaping Liao AU - Yang Lu AU - Xiaohua Wang AU - Ding Qian AU - Rujiao Cheng AU - Dongli Zhu PY - 2015/11 DA - 2015/11 TI - Wireless Graphene Nanowires QCM Ethanol Vapor Sensor Network Based on ZigBee Protocol BT - Proceedings of the 2015 International Conference on Test, Measurement and Computational Methods PB - Atlantis Press SP - 119 EP - 122 SN - 2352-538X UR - https://doi.org/10.2991/tmcm-15.2015.30 DO - https://doi.org/10.2991/tmcm-15.2015.30 ID - Liao2015/11 ER -